|Size||3 inch x 2 inch|
|Noise Level||Less than 65 DB|
|Vibration Level||Less than 2.3 micrometer|
|Power Supply||230 Volt AC|
|Gross Weight||130 KG Approx.|
|External Dimension||1020 mm L X 710 mm D X 2135 mm H|
|Internal Dimension||915 mm L X 610 mm D X 610 mm H|
Owing to our vast manufacturing facilities, we are able to manufacture and supply Bio Safety Cabinet.
These cabinets are manufactured using high quality spare parts and components following standards and norms of the industry. Owing to their perfect design and trouble free functioning, clients widely appreciate our products.
Class II Type A 2Body & Dimension:
The main body should be made in mild steel (Heavy gauge- 19 G)Table top and working zone should be made of stainless steel [(304 grade)- Heavy gauge- 20GTable top should be in two parts:
Front perforated portion 5” in sizeNon-perforated working zoneOther part details:
Work Area should be approximately 900 x 600 x 600 mm (3 – 2 feet) in size with a shutter opening of 489 mmOverall Size of the cabinet should be approximately 1100 x 825 x 2450 mmAir Flow should be vertical downflow with 70 % Surqulation 30 % exhaustCleanliness level should be less than 3.5 particles/ litre of 0.5 μm and largerAverage airflow should be 90±20 fpm (downflow)Constructed in SS 304 working table top made of Stainless steel grade of 316 (20 swg)Floor standing modelVertical laminar airflow70 % air circulation and 30 % ExhaustWork Chamber will be on Negative pressureClass 100 environmentHigh grade 0.3 micron Supply HEPA filter Efficiency 99.97%Germicidal Ultra Violet tube lightFluorescent tube light for illumination of work surfaceDynamically balanced blower-motor assemblyLow noise and vibrationAcrylic pressure ManometerAir velocity up to 100 ft/minMounted on Castor wheelsAesthetically designed and constructedStandard Accessories:
HEPA filters should be MINI PLEAT with 99.99% efficiency for 0.3 microns with integral metal guards & filter frame gaskets and manufactured in class 1000 super clean air-conditioned environment for longer life.Air pressurization system should be statically and dynamically balanced, fitted with a special vibration-reducing system to suit low noise and vibrations.Front door should be made of acrylic (4mm), adjustable as per lab requirement, vertical sliding fully auotomaticUV lamp should be in working zone (40 microwatts/ square cm at 254 nm or better) and placed so that the operator cannot see directly i.e. eyes should be always protected.Should have Rehabitable pre-filters should have efficiency of more than 80%.Should have Switches & Electrical sockets outlets for 15/5 amp.Should have Pressure Monitors like Magnehelic gauge are required to indicate pressure drop acrExhaust blower should be capable for 100% exhaust interconnected with supply air blower system should only start when the negative pressure is developedBack holes should be there on vertical walls to provide more work area on tableShould have Switches & Electrical sockets outlets for 16 amp.